FANUC PMC-MODEL SD7
EDIT DATE DESIG.
(3) ZOOM IN/ZOOM OUT OF WAVEFORM
Pushing [Z.IN] soft key magnifies the display of chart. Pushing [Z.OUT]
soft key reduces the display of chart. Pushing these soft keys also change the
scale value of the graduation on the graph. When trace is just finished, the
default zooming level was the most magnified level. In [Z.OUT] mode, gray
box is displayed as following screen example when the transitions of signal
cannot be expressed accurately enough. The limitation of [Z.OUT] displays
all of result of the trace in one page.
(4) EXCHANGE OF SAMPLING SIGNAL
Pushing [MV.UP] soft key exchanges the signal indicated by the signal
cursor for the signal one line above. Pushing [MV.DOWN] soft key
exchanges the signal indicated by the signal cursor for the signal one line
below. The result of the operation is cancelled by the execution of trace or
putting the power off. When you would like to preserve the order of
displayed signals against the executing or powering off, please change the
order on “SAMPLING ADDRESS” screen.