OPERATION12. GRAPHICS FUNCTION
4 Input numerics by numeric keys.
5 Press the
The input numerics are set by these operations and the cursor
automatically moves to the next setting items. The set data is held even
after the power is turned off.
6 Set the operation mode to the memory mode, press function key
, and call the part program which should be drawn.
7 Press function key
for a small MDI) several times to
redisplay the PATH GRAPHIC (PARAMETER) screen, then press soft
[EXEC] to display the PATH GRAPHIC (EXECUTION) screen.
* * * *
* * * * * *
10 : 10 : 40
PATH GRAPHIC (EXECUTION)
8 Press soft key [(OPRT)], then press soft key [AUTO] or [START].
[AUTO] enables automatic scaling. See item 7 in
introduction of path drawing and the description of soft key
in Explanations for details. Drawing is now started. During drawing,
the message ”
DRAWING” blinks at the lower–right corner of the CRT
9 Press soft key
[STOP] to pause drawing. The indication of “STOP”
blinks at the lower right corner on the CRT screen. Press soft key
[START] to start drawing. In addition, press soft key [REWIND] to
redraw from the top of program before pressing soft key
10 Execute the last of part program (M02/M30) to end drawing. This
will cause, blinking of the “
DRAWING” light to turn off. The
beam path view drawn can be retained until the power is turned off
unless a new beam path view is drawn.